5. Ultrasonic Transducers
The absolute calibration of a piezoelectric transducer refers to the determination of the relationship between the electrical and acoustical quantities for a transducer coupled to a solid. It is shown how a well-characterized ultrasonic system, consisting of source,structure and receiver can be used to make such measurements for transducers operating as sources or receivers. Results are given, showing the effects of coupling between transducer and solid and the electrical characteristics of the associated source and receiver electronics. This poster paper describes the principles and procedures for making absolute ultra·sonic measurements with a piezoelectric transducer coupled to a solid. Publication of these results will be in the 1979 Ultrasonics Symposium Proceedings. As described in a recent review article, the complete characterization of an ultrasonic transducer acting as source or receiver entails two parts. One part deals with the tranduction process in which the relationship between electrical and mechanical quantities is established. The second deals with the characterization of the radiation field of the transducer. This paper concerns itself with measurements which characterize the transduction process of a piezoelectric transducer coupled to a solid.
Sachse, Wolfgang, "Absolute Ultrasonic Measurements with Piezoelectric Transducers" (1980). Proceedings of the DARPA/AFML Review of Progress in Quantitative NDE, July 1978–September 1979. 21.